What you need:
- Windows based PC
- Android USB micro B cable
- Download the DfuseDemo application
- Download the automed Firmware
Download the DfuseDemo application on your Windows based PC and open the application. (click here)
Download/Save the automed Firmware onto your PC. (click here)
Connect the USB to automed USB connecter. (tip: do not have the USB connected to your PC for this step)
Press and hold the automed applicator trigger while plugging in the USB cable to the PC. Keep the trigger pulled until the countdown is complete.
The device will now display the bootloader screen and the trigger can be released.
On the DfuseDemo application; the DFU device is the automed applicator.
Note: Steps 3-5 must be completed first before the DFU Devices box will show: STM Devices in DFU Mode.
Click Choose on the DfuseDemo application and select the firmware (*.dfu) then click Open.
Click the Upgrade button.
Accept the pop up message.
The new firmware will be loading and once the DfuseDemo application says Upgrade successful! You can then click Quit.
Connect the battery to the device and ensure it powers up correctly and displays the new firmware version V3.30.
If at any time you require further assistance, please contact our support department firstname.lastname@example.org.
The following is a list of related articles about the automed “Set & Forget”